Chemical and glass thermometers for axillary temperatures: how do they compare?
Axillary temperatures recorded with a disposable chemical thermometer (DCT) measured a mean 0.29 degrees C higher than a mercury in glass thermometer (MGT) but differences could be wide. Differences between the same methods were however also wide. The DCT is safe and easy to use. Provided the higher readings are taken into consideration it is a suitable alternative to the MGT.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.